24V small peristaltic pump price is affordable. Peristaltic pump with 60 mL/min flow rate. The silicone tube or BPT tube can be chosen in our website. The pump body base is designed with strip mounting holes, which can be directly fixed to the user equipment.
Tips: Peristaltic Pump Troubleshooting
The peristaltic pump is a very important component to the proper operation of an autoanalyzer instrument. If it is not functioning properly it is possible to have problems with reproducibility, accuracy, and prematurely worn pump tubing.
First make sure that the roller heads are not excessively worn. Over time the pump tubes can wear into the material itself and cause a poor peristaltic motion as the roller head moves against the pump tubes. It is important to replace the roller heads when they are notably or significantly worn.
You might as well check the RPM of the pump to insure that the proper rotation is occurring. Sometimes pump main boards or motors will go bad and produce a slow, fast, or intermittent rotation which will significantly effect the peak shape. Simply mark a point on the roller head and measure how many rotations occur per minute.