Three phase variac variable transformer for laboratory, 20A, 15kVA rated power, input voltage 380V, variac voltage regulator can be achieved in voltage regulation, temperature control, speed control, dimming, power control, etc.
- Model: ATO-VVR-TH15
- Rated Power: 15 kVA
- Input Voltage: 380V
- Output Voltage: 0~430V, 0~500V
- Frequency: 50/60 Hz
- Rated Full Offset Current: 20A
- Insulation Resistance: ≥5 MΩ
- Pressure Resistance: 2000 V/1min
- Weight: 49 kg
- Dimension: 300mm x 240mm x 580mm
- Certificate: CE
Installation and use of three-phase variac transformer should be noted: the wiring must be firmly pressed, there can be no loose.
Three-phase variac transformer is a three-phase four-wire system, the input side should be connected to the zero line, when using the regulator to adjust the voltage should be adjusted from 0V to high voltage, the output voltage is different corresponding to the load power is different.
3 Phase 15 kVA Variac Transformer Details
Tip: Three-phase variac transformer working principle
Three phase variac transformer is a special transformer whose output and input share a set of coils. Step-up and step-down are implemented with different taps, and the voltage of the part of the tap less than that of the common coil is reduced, and the voltage of the part of the tap that is more than that of the common coil is increased. In fact, the principle is the same as that of ordinary transformers, except that its primary coil is its secondary coil. In general transformers, the primary coil on the left generates voltage through electromagnetic induction, and the secondary coil on the right is adjusted to affect itself. An autotransformer is a transformer with only one winding. When used as a step-down transformer, a part of the turns is extracted from the winding as a secondary winding. When used as a step-up transformer, the applied voltage is only applied to one part of the winding. The part of the winding that belongs to both the primary and the secondary is usually called the common winding, and the rest of the variac transformer is called the series winding. Compared with the ordinary transformer, the autotransformer of the same capacity is not only small in size, but also has high efficiency, and the capacity of the transformer is high. The larger the voltage, the higher the voltage. This advantage is more prominent, therefore, with the development of power system, the improvement of voltage level and the increase of transmission capacity.