1. Basic information
2. Signal Source Output Accuracy
Tips: Troubleshooting C512 Handheld Precision Signal Source Calibrator1. The calibrator cannot turn on or automatically shut down.Possible reasons:a. Low batteryb. Poor contact between the plug of power adapter and the socket of the generatorTroubleshooting:a. Replace the 4 AA batteries (1.5V) with new ones.b. Ensure the plug is tightly fitted to the socket.
2. The device displays “ERROR”.Possible reasons: Low batteryTroubleshooting: Replace the 4 AA batteries (1.5V) with new ones.
3. The signal output is intermittent.Possible reasons: Poor contact of the probeTroubleshooting: Ensure the probe wire and probe clip is connected well; the probe wire and the output of signal generator are tightly fitted.