Pt100 RTD sensor probes with 4 kinds of dimension can be used in different temperature measurement applications. 3 wire Pt100 probes are usually used with temperature controller, signal isolator and paperless recorder to form a temperature monitoring system.Specification:
Note: * denotes optional parameters, you can select parameters in drop-down lists to finish your order. Some customizable parameters can't be selected from the lists directly, please contact customer service staff for further communication.
Tips: RTD sensor probe for air or object surface temperature measurementCompared to the RTD sensor, thermocouples can operate at much higher temperatures and respond faster. However, the voltage signal generated by a thermocouple is low and may require additional amplification, which is a disadvantage in environments with high electrical noise. Unlike thermocouples, RTD sensors do not require reference points, ice baths, or temperature compensation circuits. RTD sensor has a very low thermal mass, so it provides true surface temperature measurement value and response times as fast as 50ms.When using an RTD sensor probe to measure the air temperature of a pipeline, the front end of the air temperature probe must be perpendicular to the measured pipeline and fully contact the measured air. The contact surface between the surface RTD probe and the measured object must be flat, otherwise, the measurement result will be affected during the measurement.